Medical Use HYZ-100C Stationary X Ray Machine
|FOB Price:||US $3,340 / Piece|
|Min. Order:||1 Piece|
|Min. Order||FOB Price|
|1 Piece||US $3,340/ Piece|
|Production Capacity:||200 Units Per Month|
|Transport Package:||Wooden Case|
|Payment Terms:||T/T, Western Union|
- Model NO.: HYZ-100C
- Type: X Ray Equipment
- Group: All People
- Medical Devices Reg./Record No.: Gsyjxz20123301226
- Certificate: CE ISO
- X-ray Tube Focus: Small Focuse:1X1mm, Big Focuse: 2X2mm
- Grid Ratio: 8:1
- Warranty: 12 Months
- Trademark: healicom
- Origin: Jiangsu, China
- Classification: Imaging Diagnostic Equipment
- Certification: CE, ISO13485
- Medical Device Regulatory Type: Type 2
- Power Voltage: Single Phase 380V/220V ± 10%
- Power Frequency: 50Hz ± 0.5Hz
- Bed Face(L*W*H): 1960mmx660X700mm
- Ray-Filter Focusing Distance: 100cm
- Place of Orign: China
- Specification: CE ISO
- HS Code: 9022140090
Filter-ray device photography
1. Single bed and single bulb.
2. Design according to the electric mechanism of middle-size machine. The function is full and reasonable.
3. The high-tension system is of march-past structure. Equipped with a full wave rectification high-voltage generator and independent X-ray tube unit, with large output power and reliable and stable operation.
4. Supply voltage V, roentgenoscopy, photography KV and stepless adjustable.
5. Equipped with the manostat for filament of X-ray tube and the space charge compensator as to ensure the stable and accurate output.
6. Equipped with the capacity limitation and protection circuit and Kv mA sinter lock limitation and protection.
7. Adopt the digital circuit timer, grade according to R10 priority coefficient and can control time exactly
8. Examining bed is of electric back rotation and is equipped with vibrating filter-ray device.
9. The photographic flat bed has a vibrating ray-filter.
10. The upright post has the bottom track only, which is easy while installation.
|Supply power||Voltage||Single phase 380V/220V ± 10%|
|Frequency||50Hz ± 0.5Hz|
|Inner Resistance||2Ω at 380Vl; 0.6Ω|
|Photograph||Max. Dc output voltage||90KV||100KV|
|Max. Dc output current||100mA||125mA|
|current||small focus 25mA and big focus 50, 75, 100, 125mA four grades together|
|time||0.04~6.3s,23 grades together|
|High Voltage Transformer Unit||capacity||7KVA||8.4KVA|
|focus||small focus 1.8mm big focus4.3mm|
|Photographic bed(YZ-III)||Bed face(L*W*H)||1960mmX660X700mm|
|Max size of cassette for photograph||356mmX432mm(14" * 17")|
|Pillar of X-ray source unit||move lengthwise (along the bottom track)||>1350mm|
|move up and down (along upright post)||720-1800mm|
MAX.CAPACITY OF PHOTOGRAY
Focus of X-ray tube
|photography mA||Max. photography KV||Max. allowable expose time|
|small focus||25mA||90KV 100KV||1.0s|
|big focus||50 mA||90KV 95KV||6.3s|
|75 mA||90KV 95KV||5s|
|100 mA||85KV 90KV||3.2s|
High-Voltage generator unit,
High voltage cable,
X-ray tube unit,
Chest film stand
Manual Diagnostic Table