Hot Sale HV-400A Hospital Ventilator Price
|FOB Price:||US $2,250 / Piece|
|Min. Order:||1 Piece|
|Min. Order||FOB Price|
|1 Piece||US $2,250/ Piece|
|Production Capacity:||1000 Units/Month|
|Payment Terms:||T/T, Western Union|
- Model NO.: HV-400A
- Kind: Surgical Equipment
- Group: All People
- Medical Device Regulatory Type: Type 3
- Certificate: CE ISO
- Gas Requirement: Medical Oxygen and Nitrous Oxide
- Modes of Ventilation: A/C, Ippv, Sippv, Imv, Simv, Peet. Manual,Sigh etc
- Peep: 0.1 ~ 1.0 Kpa
- Imv Frequency: 1~20 Bpm
- Trademark: healicom
- Specification: CE ISO
- HS Code: 9019200000
- Type: Ventilator
- Certification: CE, ISO13485
- LCD Display: With LCD Display
- Medical Devices Reg./Record No.: Gsyjxz20143540786
- Display: 10.4′′ LCD Display
- Safety Valve: <12.5 Kpa
- Ventilator Compliance: ≤30 Ml/Kpa
- Ptr: -10 ~ 10 Hpa
- I/E Ratio: 4:1~1:4, Has The Inverse Ratio Ventilation
- Transport Package: Carton
- Origin: Jiangsu, China
The HV-400A ventilator is gas driven by medical compressed oxygen and compressed air. In the inspiratory phase, two streams of compressed gas (compressed oxygen and compressed air) flow into a high performance air-oxygen mixer to form a mix of oxygen and air with a certain pressure.
The automatic conversion for standby power while AC power outage, keep working not less than 1 hour.
Display: 10.4''LCD Display Following Parameters & Real time waveform display
Operating mode and respiratory rate;
--Ventilator control frequency set by medical staff;
--Tidal volume of each breath in the patient;
--Occurrence of spontaneous breath in the patient and spontaneous breathing frequency;
--Expiratory phase and inspiratory phase and actual respiratory rate;
--Setting and adjustment of inspiration triggering pressure, PEEP, and highest airway pressure;
--Real-time variation in internal airway pressure.
--Environmental Protection Material construction, hardly to be broken.
--Trolley: Fitted with 4 nos anti-static rubber castors; two of which are lockable for braking and easy maneuverability with foot operated brake provisions
10.4'' LCD Display
pneumatically powered and electrically controlled system
Fitted with 4 nos anti-static rubber castors; two of which are lockable for braking and easy maneuverability with foot operated brake provisions
Medical oxygen and nitrous oxide with a pressure ranging from 280~600kpa
1 ~ 99 bpm
Maximum Minute ventilation
≥18L / min
modes of ventilation
A/C, IPPV, SIPPV, IMV, SIMV, PEET. MANUAL, SIGH, SPONT
0 ~ 1.0 kPa
-10 ~ 10 hPa
4:1~1:4, Has the inverse ratio ventilation
Setting Range: 50~1500ML Display Range: 0~2000ml ; resolution ratio: 10ml
Maximum Safety Pressure
one deep breath per 80 controlled respirations, the inspiration time is 1.5 times of the setting point
Pressure limit range
Airway pressure alarm: Audible and visual and with yellow and red color indicating
Lower: 0.1 kPa ~ 6.0 kPa; Upper: 0 ~ 5.0 kPa
Ventilation Volume alarm
≥ 18L/ MIN
Power Supply Alarm
Ac/dc power supply are after failing to send out alarm immediately
Alarm Time: keep >120s
10 ~ 40oC
no higher than 80 %
860 hPa ~ 1060 hPa
100-120 Vac, 50/60 Hz;
Attention: the AC power supply used for the ventilator machine must be well grounded.
-15oC ~ +50oC
no higher than 95 %
86 kPa ~ 106 kPa.
It should be stored in a room without corrosive gas and well ventilated
comply with the requirement of GB/T 15464
Between the packaging box and product, soft material with appropriate thickness provided to prevent loosening and mutual friction during transportation
Moisture protection and rain protection to ensure that product is protected from natural damage.
Safty & Alarm
Gas source pressure
Immediate alarm; "high level alarm" indicator flickers raising an audible alarm.
Airway Pressure Alarm
The airway pressure exceeds 15hpa. The maximum delay is no longer than 17s. The "high level alarm" indicator flickers raising an audible alarm.
Tidal Volume Alarm
Lower than 50ml, The alarm will be raised after one breathing cycle.
Ventilation Volume Alarm
Up: 18L/ Min
Battery Voltage Alarm
It alrms when the battery voltage is lower than 10.5V.
Power On Self Test
Self test before working, assure it's working normally.
A leather bag ( Blue)
Tools with the machine
User Manual (English Version)